美国NDT James C-CL-3000 Chlorimeter混凝土氯离子含量测定仪
The Chlorimeter Chloride Test System
A field kit for the determination of chloride ion content in concrete, fresh cement, masonry, most other construction materials, and water.
Service life prediction
Features & Benefits
Fast - Results within minutes at the site.
Economical - Low cost per sample compared to laboratory testing.
Accurate - Results are comparable to laboratory testing.
Covers wide range from 0.002% to 2% chloride by weight
Internal Memory to store readings for later upload to a PC via USB.
Digital display for direct reading of percentage of chloride by weight.
Conforms to AASHTO-T- 260.
Menu in English and Spanish.
The Chlorimeter produces results on-site, within minutes that are accurate and comparable to expensive laboratory tests. It measures the electrochemical reaction of a weighted sample placed in an extraction liquid. It automatically shows a temperature compensated reading of percent of chlorides on its digital display. A wide range - from 0.002 to 2% chloride by weight - is covered.
The determination of the chloride ion concentration in concrete is essential in assessing the need for maintenance on, for example, bridge decks and parking structures. The test can also be used to ensure that materials used in new construction are free from potentially harmful chloride ion levels.
With this method, the concentration of acid soluble chlorides is measured. In most cases, this is equivalent to total chloride concentration.
A sample of powder is obtained by drilling and careful quartering. Then an accurately weighed 3 gr. (0.1 oz) sample is dissolved in 20 ml (0.67 fl. oz.) of extraction liquid which consists of a precise, measured concentration of acid. For sampling wet concrete a 3 gr. (0.1 oz.) sample of mortar (i.e. without coarse aggregate) is used. Typically for concrete, samples that are to be analyzed for chloride are obtained at various depths, thereby supplying the engineer with a chloride ion profile.
The chloride ions react with the acid of the extraction liquid in an electrochemical reaction. An electrode, with integral temperature sensor, is inserted into the liquid and the electrochemical reaction measured.
A uniquely designed instrument converts the voltage generated by the chloride concentration and sensor into a reading for the user. The instrument automatically applies the temperature correction and it shows the chloride concentration on a LCD display percentage by weight. The unit uses 4 'AA' batteries for power. The Chlorimeter can also store test results and upload them to PC via USB.
Once the sample is obtained, test results can be determined and read in less than five minutes. These initial values give a valuable indicator of the current chloride concentration values. Accurate results are typically obtained after the sample has been in the extraction liquid for 24 hours.
To avoid contamination, the electrode should be thoroughly washed with de-ionized water after each test.
Replacement packs, containing twelve bottles of extraction liquid, each for one time use, are available. Five calibration liquids each with known concentrations are supplied with each pack. Bulk packs of one hundred extraction liquids are also available for larger testing jobs.
Thecalibration liquids are used to establish the calibration curve, and to check that the system is functioning correctly. Calibration is not required for each use. The calibration liquids are colored to avoid confusion between them and with the extraction liquid.
All equipment necessary to complete the chloride test is supplied in a standard size carrying case.
Chloride combination electrode with externally mounted temperature sensor, cable, and connectors.
Battery powered, high impedance, electronic meter, with temperature compensation circuits and microprocessor for direct conversion to percentage of chloride. Uses four 'AA" size batteries.
Optional pack of 12 jars each with 20 ml (0.67 fl. oz.) of extraction liquid and 5 jars of colored calibration liquid.
Bottle of electrode wetting agent.
Optional bulk pack of 100 jars extraction liquid and 20 jars of colored calibration liquid.
美国NDT James C-CL-2000 Chlorimeter混凝土氯离子含量测定仪
特 性 :
『 含氯量以百分比和 ppm 表示，可直接读数。
用这种方法可以测出混凝土内酸溶性氯化物的含量，其单位为重量百分比和 ppm 浓度。在大多数情况下，上述数值与总含氯量是相当的。『 在混凝土构筑物上，用钻孔法取样，将钻出的 3 克混凝土粉溶入 20 毫升的酸性萃取液内，萃取液的浓度是经过精确标定的。混凝土中的氯离子与萃取液中的酸发生电化学反应。
『 仪器带有一盒萃取液，内装 12 瓶，每次测试用一瓶。盒内还有两瓶校准液，每瓶校准液的浓度已预先标定，其用途是使操作者能校准仪器的精度。为避免在使用中发生混淆，校准液有两种颜色，一瓶是红色的，一瓶是蓝色的。
C-CL-2012 备件盒。内有塑料瓶装萃取液 12 瓶，每瓶 20ml ，盒内还有 5 瓶有颜色的校准液
以上是美国NDT James C-CL-3000 Chlorimeter混凝土氯离子含量测定仪的详细信息，如果您对美国NDT James C-CL-3000 Chlorimeter混凝土氯离子含量测定仪价格、厂家、品牌、型号、图片、规范、规格、原理有疑问，或需要详细报价单、宣传册、中英文产品说明书，请点击我公司网站www.